The TECO 42C User's Manual is now available to ARB staff.
The objective of the Air Monitoring Web Manual is to provide standard procedures for maintaining and operating air monitoring stations and detailed instructions for acceptance testing, maintaining, troubleshooting and calibrating specific analyzers or support equipment.
The accuracy and/or validity of data obtained from any analyzer depends upon analyzer performance and operator proficiency. Deviations from the recommended procedures set forth in this manual, as well as in the Manufacturer's Instruction Manual, may result in the collection of invalid data.
Therefore, the station operator must become familiar with the information contained in this manual as well as each Manufacturer's Instruction Manual in order to achieve a minimum level of competence and acceptable level of data quality.
For questions or comments related to these documents, please email the Operations and Data Support Section.
Note: To print an appendix, first download the file to your computer, then open the document and choose print.
|Number & Type||Document Name||Date Posted||File Size|
|100 chk||TECO 42 Oxides of Nitrogen Analyzer||03/28/07||54.9 KB|
|101 chk||Advanced Pollution Instrumentation, Inc (API) Model 200 Oxides of Nitrogen Analyzer||03/28/07||49.6 KB|
|102 chk||Advanced Pollution Instrumentation, Inc (API) Model 200A and 200E Oxides of Nitrogen Analyzers||08/28/09||14.9 KB|
|102 chk||Advanced Pollution Instrumentation, Inc (API) Model T200U Oxides of Nitrogen Analyzer||11/27/13||41.4 KB|